What does RGT/SMP Error on the CA610 Fluoride analyser mean?
Displaying RGT/SMP Error on the CA610 Fluoride analyser.
RGT/SMP Error is on software Ver 1.4 and can be cleared on the CA610 Fluoride analyser. It may be caused by bubbles in the sample stream, or that cling inside the measurement chamber. The message doesn't happen as often with the latest Ver 1.5 software. If this is a reoccurring problem, and the instrument has Ver 1.4, the entire circuit board has to be replaced in order for it to update to the newest version.