How much flow do I need to send to my BioTector Total Organic Carbon (TOC) Analyser?
BioTector TOC (B3500 and B7000) flow requirements.
Sample needs to be pulled from an ambient pressure source with the BioTector TOC analyser. If using the Hach Sample Overflow Chamber, 19-BAS-031, we want to have 250-300 mL/min of flow. If a user creates their own sample vessel, we want that vessel to have enough flow to replace the contents 8 times per sample cycle to ensure proper analysis. Sample times are analyser specific and can be checked within the menu.