Why is does the DR 6000 software require the multi cell adapter on the 10mm size on HR Chlorine program 88, but the Hach procedure requires a 1 inch sample cell?
DR 6000 high range (HR) chlorine program 88 requires the 1 inch sample cell.
Older software versions had this message in error. The 1 inch sample cell is correct for this program. Updating the software version will eliminate this error.