What has happened if the HQd meter started playing music and now does not work?
HQd meter started playing music and stopped working
Typically this means a fatal error on the instrument has occurred. This can happen when attempting to transfer data or update the meter with a USB stick that is too large. Use a USB stick that is 2GB or smaller with the meter. If this error happens, the meter must be sent to Hach Service for repair.