What data can be pulled from the 2200 PCX particle counter with UDG1000 software?
2200 PCX data and reporting with UDG1000 (Universal Data Gateway) software.
Currently when using a 2200 PCX with UDG1000 software the local display of the particle counter will show the raw count of particle size set up by the CRTS software. This is independent of the configuration of the 2200 PCX. In UDG the particle counter will show the raw count as well. A calculation must be made to determine the normalized (particles per mL) in a program such as Microsoft Excel. There are updates for the UDG1000 firmware that will increase the capabilities in the near future. A release date has not been set for these updates.