What type of water can I analyze with the QBD1200?
QBD1200 and process water
The QBD1200 is designed to be a clean water TOC analyzer. This means clean drinking water, post treatment waste water, lab grade water, or pharmaceutical water is best. The unit reads up to 100 ppm and the sample should not have any visible particles. If you can see particles in the water, filter with a .45 micron filter. The unit should not be run with aggressive acids or bases, chlorides above 300 ppm, or with any foamy or bubbly solutions.