What should the Phi0 (T) value be for a good zero calibration on an Orbisphere K/M1100 high range sensor?
Orbisphere and different calibration parameters compared to low range sensor.
In the MAIN>SERVICES>DIAGNOSTIC>SENSOR>ZERO DIAGNOSTIC View, Select Show Criteria. Readings should fall between 50-62°. This varies from the K/M1100 low range sensors Phi0 (T) range of 60-75° shown in the Service Manual.