How do you reset the profile counter on the UVAS sc sensor?
UVAS and reset counter
When you change the profile, you should go in the Sensor Setup Menu > Select the sensor >Test/Maint >Maint. Proc. >Replace profile. The wiper will then be driven out so you can replace the profile and display "Replace Profile!". Once you have replace the profile (wiper) press Enter. The wiper will go back in and will take you back to the Test/Maint menu, press the back key, controller will display "Return Probe to Process", press enter. "Please wait" and a status bar will be displayed. When it is done you will be taken back to the main measure screen and the counter will be reset.