My resistivity is above 18.2MOhm or I am getting "High Resistivity." How can I resolve this?
A1000 and High Resistivity
This can indicate air a in the measurement cell. Make sure the unit is upright when running and there are no air bubbles in the lines. Send the instrument in for calibration if installation is correct per the user manual.