What does "Sensor Warning" mean when displayed on a CL17, and what are the corrective measures?
CL17 and Sensor Warning
This message indicates that the analyser has detected something wrong using its internal diagnostics. To see what the error is, push the menu key. Push the down arrow key until you reach the alarm menu, then press enter. Press the down arrow until 'Recall Warning' is displayed, then press enter. An error message will be displayed. Look in the instrument manual to find the error message and corresponding correction.