What are the pressure/flow requirements if customer supplies their own air source for the air blast?
Differential pH sensor and Air Blast/Washer Head Requirements
The standard pHD Air Blast compressor normally supplied with the pHD Air Blast System provides a maximum intermittent pressure of 20 psi, at 0.17 cubic feet per minute (CFM). The same air/flow requirements apply to the Air Blast/Washer heads for the 1.5” Encapsulated pH/ORP sensors. Providing a pressure higher than 20 psi could cause damage to the sensor.