What is the optimal sample flow rate for the CL17?
CL17 and Flow Rate Requirements
The optimal flow rate for the CL17 is 300 mL/min. To check the flow rate, take the sample line off the plastic elbow fitting below the bottom left corner of the pinch plate assembly. Measure the sample water in a beaker from this line for 1 minute and adjust the flow rate at the ball valve of the sample conditioning kit until the optimal flow rate is achieved.