The stabilization time on the CDC401 probe is slow, what could be causing this issue and how can it be resolve?
CDC401 slow stabilization
Slow stabilization time can be caused by:
- Mineral or sample buildup on electrodes which can be resolved by cleaning the probe (refer to the User Manual for Conductivity Probe Models: CDC40101, CDC40103, CDC40105, CDC40110, CDC40115 or CDC40130).
- Bubbles trapped under the probe tip which can be resolved by gently shaking the probe side to side to dislodge any trapped air bubbles.